New to Ham Radio?
DX Cluster Spots
eHam Help (FAQ)
Support the site
The eHam Team
Show all reviews of the SARK-110 Vector Impedance Antenna Analyzer
You can write your own review
of the SARK-110 Vector Impedance Antenna Analyzer.
Aug 5, 2013 02:27
Send this review to a friend!
Time owned: 3 to 6 months
Opened the box and first looks very nice good feel to the touch
Looked nice quality
Tested on know antenna ... Guess what total rubbish data results .... Suspected connector that came with it so ordered
A new mcx plug that arrived and made a nylon jig to hold everything still and I then got mixed results some rubbish data and some good data ..
So what do I do now ship it back to china or take a look inside
They provide the small hex key.
And under the magnifier I found the mcx connector was dry
Resolderd and then retested. .........
Perfect Please note I think this is the wrong type of connector for this machine ..
But to be honest it excellent .. I mean really nice
Fantastic even just. A bit small needs to be in a protective case option or your going to break it in the field because it's so light. You also need a much longer micro cable than it comes with. But that's all I can say bad about the sark110
So buy it but watch that socket.
Product is in production.
You are not subscribed to this review.
If you have any questions, problems, or suggestions about Reviews
please email your Reviews Manager